|Rotor Capacity||6 x 0.2mL tubes|
|Min Centrifugal Force||500 x g|
|Max Centrifugal Force||8,000 x g|
Time Spin of up to 60 minutes
Before Use: Balancing
Before starting the centrifuge, make sure that your samples are placed opposite each other on the rotor.
Also check that your samples are the same weight.
Using the Centrifuge
To operate the centrifuge, select the module from the status screen.
There are two modes of usage; the Burst Spin and the Time Spin. The Burst Spin is used for short spins with no pre-deﬁned time. This mode is usually used when you need to run the centrifuge for a number of seconds, such as when collecting samples at the bottom of the tube. The Time Spin is used for spins at user-deﬁned speeds for a pre-deﬁned time.
The lid has to be closed to select either Burst Spin or Time Spin options.
To use the Burst Spin option, select the circular arrow button using the orange click-dial. Hold down the orange click-dial and the centrifuge will start spinning. A counter will record how many seconds the centrifuge has been active. Once the user lets go of the orange button, the centrifuge will slow down and come to a halt. The lid automatically unlocks once the motor has stopped and it is safe to open. Wait until you hear a click from the lid.To use the Time Spin option, select the clock symbol using the orange click-dial. Set the duration of the spin and the G force required.
Please note that some tubes, even though they physically ﬁt into the centrifuge rotor, will bend too far at high speeds and can damage the centrifuge. These include some skirted tubes.
Intention of use
The centrifuge is intended for the separation of suspensions and solutions held in the appropriate 1.5ml.
- Only operate the centrifuge after you have read and understood the manual.
- It is the responsibility of the Bento Lab owner to ensure that all persons using the centrifuge have been trained and that full records of training are maintained.
- Wear the necessary protective equipment and follow good lab practices, including for the storage and disposal of samples and tubes.
- Irrespective of the organism, chemical or reagent which you are using, always ensure you are handling, storing and disposing of the material as per supplier instructions and as per local and national regulations.
- Do not use the Bento Lab if there are exposed wires, signs of severe mechanical damage or corrosions.
Warnings to avoid personal injury, exposing others to risk, damage to centrifuge and rotor
Risk of ﬂying debris and/or ﬁre from mechanical failure of rotor
- Prior to use, inspect the centrifuge visually and ensure that power cables, lid and seals are clean and in good condition and that no wires are exposed.
- Stop the centrifuge immediately if an unusual condition (noise or vibration) begins.
Risk of skin irritation from opening the centrifuge and touching the rotor during operation
- Do not open the cover when the rotor is still spinning.
- Do not attempt to slow down the spinning rotor by hand.
Risk of cuts and infection from broken sample tubes
- Only use sample tubes of size 0.2ml or 1.5ml
- Wear suitable protective gloves.
- Disinfect the centrifuge if there are any spills. If necessary, clean the surrounding area.
- Decontaminate and dispose of the broken sample tubes as necessary.
- If you cut yourself, seek ﬁrst-aid.
Risk of contamination from use of hazardous samples
- Only use pathogenic, toxic or radioactive substances if safety precautions listed on the product of interest are observed.
- Handle and dispose of the samples as per good laboratory practice procedures.
Risk of rotor damage from imbalance of samples in centrifuge during operation
- Load the samples in the centrifuge evenly.
- Ensure they have the same mass.
- Ensure the centrifuge is placed on a level surface.
Risk of ﬁre, combustion from operation of centrifuge in vicinity of ﬂammable or combustible vapours which can ignite inside the centrifuge
- Work in a well-ventilated space
Risk of ﬁre, contamination or corrosion from sample leaks
- Check tubes for cracks and deformities before use.
- Do not ﬁll tubes above the maximum recommended by manufacturer.
- Material loaded into sample tubes must be compatible with the sample tube material
- Clean spillages immediately. Spills of hazardous material must be adequately decontaminated.# The remainder of the liquid will ﬂow through. The centrifuge is insulated from the remainder of the equipment and so should not cause any shortages.
- Do not centrifuge samples which can develop ﬂammable or explosive vapours.